# Enthalpy (T)

### Measurement details

Measured with | TA DSC2500 |

By: | Fraunhofer ISE |

Additional Comments | The crucible has a hole in the lid. The sample (with crucible) was previously measured in a TGA to determine the dry mass. The sample was sufficiently dry. |

- Heating rate: 0.5 [K/min]
- Cooling rate: 0.5 [K/min]
- Sample mass: 9.92 [mg]
- Heat of fusion: 176.2 [kJ/kg] - deviation: 0.2 [kJ/kg]
- Integration range, lower limit: 295.0 [°C]
- Integration range, upper limit: 310.0 [°C]
- Onset temperature: 306.8 [°C] - deviation: 0.03 [°C]
- Offset temperature: [°C] - deviation: [°C]

- Crystallization enthalpy: 176.6 [kJ/kg] -
deviation: 0.1 [kJ/kg]
- Integration range, lower limit: 295.0 [°C]
- Integration range, upper limit: 303.0 [°C]
- Onset: 300.0 [°C] - deviation: 1.0 [°C]
- Offset: [°C] - deviation: [°C]